<H3>Special Instructions</H3><p> <p>This test is run by Image Display systems which support any of the cardiology options in the Evidence Documents profile:</p> <ul> <ul> <li>Cath Evidence option</li> <li>Echo Evidence option</li> <li>CTA/MRA Evidence option</li> <li>Stress Test Evidence option </li> </ul> </ul> <p>In order to run this test, Evidence Creator actors which support these options must complete the tests in which they store their evidence documents (SRs) to Image Manager systems.&nbsp; These tests are named ED_Creator_PACS_*_Option.</p> </p><H3>Description</H3><p>In this test, the Image Display demonstrates its ability to cardiology evidence documents (SRs) created by various Evidence Creator systems.</p><p> The Image Display must be able to render and display all Templates for the SOP Classes which it accepts.</p><p> The Image Display system should drive this test.&nbsp; The representative of the Image&nbsp; Display system should identify an Evidence Creator system which supports the same cardiology option(s) as its system. Ask the Evidence Creator vendor representative which Image Manager/Image Archive it has stored its evidence documents (SRs) to.</p><p> Note 1: The Evidence Creator is a required actor in this test.&nbsp; It has&nbsp; no transactions to perform, but will be identified so that it can be given credit for interoperability with the Image Display who renders its SRs.</p><p> Note 2:&nbsp; The Image Manager system is identified as &lsquo;optional&rsquo; for this test because it is needed to help out with Image Display system, allowing it to Query/Retrieve SRs from its system.&nbsp; Image Managers are not required to complete&nbsp; 3 instances of this test.</p><H3>Evaluation</H3><p>FOR ALL CARDIOLOGY OPTIONS (Cath Evidence, Echo Evidence, CTA/MRA Evidence, Stress Test Evidence:</p><ul> <li>The Connectathon Monitor should observe &lsquo;live&rsquo; the ability of the Image Display to query and retrieve SRs from an Image Manager system (not the Central Archive).</li> <li>The Connectathon Monitor should observe the Image Display and its ability to interpret/render the Evidence Documents from the Evidence Creator. All SOP Classes and Templates which the Evidence Creator supports should be displayed.</li> <li>The Connectathon Monitor should observe the original report at the Evidence Creator (the creator of the SR), the Image Manager, or another Image Display and spot check the Image Display/Report creator to verify the contents of the Structured Report are displayed <u>correctly</u> and <u>completely</u> </li> <li>The Image Display must demonstrate its ability to show abnormal measurements with &ldquo;readily visible emphasis&rdquo;, (e.g., font, bold, text or background color, specialized window area, etc.).</li> </ul><p>For the <strong>CTA/MRA Evidence </strong>option:</p><ul> <li>The Image Display must demonstrate that it can display a hyperlink to a referenced object for a content item with value type of IMAGE.</li> <li>The Image Display must demonstrate that it can display a hyperlink to a referenced object with rendering of spatial coordinates for a content item with value type of SCOORD.</li> </ul><p>For the <strong>Stress Test Evidence </strong>option:</p><ul> <li>The Image Display must demonstrate that it can display a hyperlink to a referenced object for a content item with value type of IMAGE.</li> <li>The Image Display must demonstrate that it can display a hyperlink to a referenced object with rendering of temporal or spatial coordinates for a content item with value type of TCOORD or SCOORD.</li> <li>The Image Display actor must demonstrates user control for rendering of ECG ST elevation or depression measurements either in units of mV or in units of mm (based on conventional scaling of 0.1 mV/mm). </li> <li>For SR documents encoded using Template x3300, the Image Display must render the contents of the Template x3304 Stress Test Measurement Groups (subsidiary to a Template x3303 Stress Test Phase Data CONTAINER) <strong><u>in a tabular format</u></strong>. </li> </ul> 