<H3>Special Instructions</H3><p> <p>The Image Display may use the Connectathon CENTRAL_ARCHIVE as a source of different vendors' waveforms, images and reports.</p> <p>Image Display please use the <em><a href=system_con_object.php>Objects Information</a></em> page to find out what your peers have uploaded on the central archive </p> </p> <H3>Description</H3> <p>The test verifies that the Image Display can retrieve and display a variety of echo image sets, DICOM SRs and Procedure Logs from various vendors. </p> <H3>Evaluation</H3> <p>On the <strong>Image Display</strong>  </p> <ul> <li>verify that images and reports from each Modality vendor can be accurately stored and displayed. The Connectathon Monitor may ask a representative of the originating Modality to verify the accuracy of the rendering of its study on the Image Display.</li> <li>If the Image Display supports the STRESS ECHO option, verify:   <ul>      <li>Each waveform is labeled with the lead name.</li>      <li>The following values must be displayed:       <ul> 	     <li>Protocol Name: either the value of (0018,1030), or the value of Code Meaning from the Performed Protocol Code Sequence (0040,0260)</li>          <li>Number of Stages</li>          <li>Number of Views in Stage</li>          <li>Code Meaning from the Stage Code Sequence (0040,0102)</li>          <li>Code Meaning from the View Code Sequence (0054,0220)</li>          <li>Stage Number (0008,2122), numbering starting with 1</li>          <li>View Number (0008,2128), numbering starting with 1</li>        </ul> 	 </li> </ul> </li> <li>The Connectathon Monitor should observe its ability to interpret/render the objects (derived images, SRs, or measurements, ie Evidence Documents) from the Evidence Creator actors. The Connectathon Monitor may ask a representative of the originating Evidence Creator to verify the accuracy of the rendering of its data, or observe the originals at the Evidence Creator and spot-check that they are displayed correctly and completely. </li>  </ul> 