<H3>Special Instructions</H3><p> <p>This tests a DOF/DOC pair that implements the Subscribe to Patient Data (SPD) option. It    requires a cooperating DOR to send data to the DOF if the DOF is not contained within the same    system as the DOF. All DOR/DOC tests should be completed before testing the SPD option.</p> <p>There are two variations of this test; be sure you select the right one:</p> <ul>     <li>DEC_SPD_By_Location _FC (this is the test described here)</li>     <ul>      <li>DOR and DOF are contained in the same system; the DOR has succeeded with the DOR/DOC tests      where no subscription was required</li>     </ul>     <li>DEC_SPD_By_Location _RFC</li>     <ul>      <li>Standalone DOF, input to the DOF requires the DOF to act as a DOC, will need a DOR as      partner; the DOF has succeeded in testing as a DOC with a DOR to assure it can receive PCD-01      messages. </li>     </ul> </ul> <p>This tests the DOF&rsquo;s ability to receive and respond to a subscription request with the    appropriate filter and send the requested filtered data. It also tests the ability of the DOC to    subscribe and receive messages containing this filtered data. The test requires the DOC to    request multiple filtered messages containing data with multiple physiological parameters for two    or more patients (or only a single patient if multiple patients are not supported) and the DOF to    provide the data requested. If the DOF or DOR supports only Vital Signs Monitors (VSMs) , there    will be a single transmission with data from two or more patients in that transmission .</p> <p>Note 1: Test names are the same for DOFs and DOCs. They play partner roles. DOFs are    responsible for coordinating with their partners in advance of the Connectathon to assure that    the DOC needed is able to send or receive the messages.</p> <p>Note 2: Systems may use validated or unvalidated data</p> </p><H3>Description</H3><p>This test utilizes subscriptions and filtering based upon location. A DOC requests data as    described below. A DOF sends a series of messages in response. For efficiency, this test may be    conducted with multiple DOCs if supported, and if desired. </p><p> If multiple patients are supported by the DOF, this test will be conducted with two patients.    If either system does not support multiple patients, you must make this known to your test    partners and the project manager. If testing a DOF that supports only VSMs, please note the    differences below. </p><p>Messages include at least three physiological parameters for each type of system supported    (e.g., if the systems support bedside monitor and ventilator data, at least three parameters are    required for each). Parameters are selected at the discretion of the participating systems. The    DOF and the DOC will participate in sending or receiving messages for a variety of supported    parameter types; for this test, the DOF will send at least two messages at each step of this test    unless the test is solely for VSM data. The steps are:</p><ol>     <li>If the DOC is receiving messages it first sends a request to terminate all transmissions.</li>     <li>DOC (or DOCs) request messages for all available parameters for the patients in the table     below at a rate acceptable to the participating systems. At least one message is received for     each patient before proceeding. Patients are individualized for the DOF.</li>     <table border="1" cellspacing="0" cellpadding="0">      <tr>       <td width="120" valign="top">        <p>System</p>       </td>      <td width="90" valign="top">        <p>MRN</p>       </td>      <td width="60" valign="top">        <p>First Name</p>       </td>      <td width="66" valign="top">        <p>Last Name</p>       </td>      <td width="142" valign="top">        <p>Location</p>       </td>      <td width="79" valign="top">        <p>Mother&rsquo;s Maiden Name</p>       </td>      <td width="73" valign="top">        <p>DOB</p>       </td>      </tr>      <tr>       <td width="120" valign="top">        <p>&nbsp; B Braun (DOF)</p>       </td>      <td width="90" valign="top">        <p>BB2008001</p>       </td>      <td width="60" valign="top">        <p>Albert</p>       </td>      <td width="66" valign="top">        <p>Brooks</p>       </td>      <td width="142" valign="top">        <p>3 West ICU - 31-1</p>       </td>      <td width="79" valign="top">        <p>Coburn</p>       </td>      <td width="73" valign="top">        <p>1/1/61</p>       </td>      </tr>      <tr>       <td width="120" valign="top">        <p>&nbsp; B Braun (DOF)</p>       </td>      <td width="90" valign="top">        <p>BB2008007</p>       </td>      <td width="60" valign="top">        <p>Irma </p>       </td>      <td width="66" valign="top">        <p>Kwan</p>       </td>      <td width="142" valign="top">        <p>Emergency - ED4-2</p>       </td>      <td width="79" valign="top">        <p>Lee</p>       </td>      <td width="73" valign="top">        <p>1/1/62</p>       </td>      </tr>      <tr>       <td width="120" valign="top">        <p>&nbsp; Philips (DOF)</p>       </td>      <td width="90" valign="top">        <p>PH2008001</p>       </td>      <td width="60" valign="top">        <p>Adrian</p>       </td>      <td width="66" valign="top">        <p>Brooks</p>       </td>      <td width="142" valign="top">        <p>3 West ICU &ndash; 34-1</p>       </td>      <td width="79" valign="top">        <p>Coburn</p>       </td>      <td width="73" valign="top">        <p>1/1/64</p>       </td>      </tr>      <tr>       <td width="120" valign="top">        <p>&nbsp; Philips (DOF)</p>       </td>      <td width="90" valign="top">        <p>BB2008007</p>       </td>      <td width="60" valign="top">        <p>Ira</p>       </td>      <td width="66" valign="top">        <p>Kwan</p>       </td>      <td width="142" valign="top">        <p>Emergency - ED4-2</p>       </td>      <td width="79" valign="top">        <p>Lee</p>       </td>      <td width="73" valign="top">        <p>1/1/65</p>       </td>      </tr>      <tr>       <td width="120" valign="top">        <p>&nbsp; Spacelabs (DOF)</p>       </td>      <td width="90" valign="top">        <p>SP2008001</p>       </td>      <td width="60" valign="top">        <p>Arthur</p>       </td>      <td width="66" valign="top">        <p>Brooks</p>       </td>      <td width="142" valign="top">        <p>3 West ICU &ndash; 35-1</p>       </td>      <td width="79" valign="top">        <p>Coburn</p>       </td>      <td width="73" valign="top">        <p>1/1/65</p>       </td>      </tr>      <tr>       <td width="120" valign="top">        <p>&nbsp; Spacelabs (DOF)</p>       </td>      <td width="90" valign="top">        <p>BB2008007</p>       </td>      <td width="60" valign="top">        <p>Ima</p>       </td>      <td width="66" valign="top">        <p>Kwan</p>       </td>      <td width="142" valign="top">        <p>Emergency - ED4-2</p>       </td>      <td width="79" valign="top">        <p>Lee</p>       </td>      <td width="73" valign="top">        <p>1/1/66</p>       </td>      </tr>     </table>     <li>DOC #1 requests messages for the patient in the ICU and for the patient in the ER     corresponding to the DOF company&rsquo;s name. DOC #2 (if participating) does the same. The DOCs may     (but are not required to) request filtering on parameters to reduce the amount of data).</li>     <li>If the DOF will transmit VSM data, then one occurrence of that data is sent in addition to     any other data. The VSM will obtain data for two patients, each at least one minute apart in     time. The VSM will provide data to the DOR and combined DOF, in one connection, with data for     one patient sent immediately following data for the preceding patient. NOTE: The VSM will use     two patients on the ICU, and the subscription for the data will be for all patients on the ICU.</li>     <li>DOC(s) terminate the subscriptions. </li> </ol><H3>Evaluation</H3><p>Project manager or monitor will observe the displays provided by the DOF and DOC. The    physiologic parameters and their measurement values, including the time of parameter measurement    will be compared. The type of display/documentation by the DOF and DOC is not constrained, it    need only be clinically useful.</p><p>Success: If the patient, the minimum number of parameters, measurements and time of measurement    sent by the DOF matches those received by the DOC in each of the three steps above, that will    indicate the test has been successful. </p> 