<H3>Special Instructions</H3><p> <p>This tests a DOF/DOC pair that implements the  Subscribe to Patient Data (SPD) option. It requires a cooperating DOR to send  data to the DOF if the DOF is not contained within the same system as the DOF. All  DOR/DOC tests should be completed before testing the SPD option. This test is  NOT performed with data from Vital Signs Monitors.</p> <p>There  are two variations of this test; be sure you select the right one:</p> <ul>     <li>DEC_SPD_Snapshot _FC</li> </ul> <blockquote>     <p>DOR and DOF are contained in the  same system; the DOR has succeeded with the DOR/DOC tests where no subscription  was required</p> </blockquote> <ul>     <li>DEC_SPD_Snapshot _RFC</li> </ul> <blockquote>     <p>Standalone DOF, input to the DOF  requires the DOF to act as a DOC, will need a DOR as partner; the DOF has succeeded  in testing as a DOC with a DOR to assure it can receive PCD-01 messages.&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; </p> </blockquote> <p>This tests the  DOF&rsquo;s ability to receive and respond to a subscription request with the  appropriate filter and send the requested filtered data. It also tests the  ability of the DOC to subscribe and receive messages containing this filtered  data. The test requires the DOC to request multiple filtered messages  containing data with multiple physiological parameters for two or more patients  (or only a single patient if multiple patients are not supported) and the DOF  to provide the data requested. </p> <p>NOTE:  Test names are the same for DOFs and DOCs. They play partner roles. DOFs are  responsible for coordinating with their partners <em>in advance of the Connectathon</em> to assure that the DOC needed is  able to send or receive the messages.</p> <p>NOTE: Systems may use validated or unvalidated  data.</p> </p><H3>Description</H3><p>This test utilizes subscriptions and filtering to  achieve a single snapshot of data for each selected patient. </p><p>  The snapshot should include sampled continuous  data (e.g., heart rate at one value a minute) and data that is intermittent  (e.g., non-invasive blood pressure).</p><p>This test should precede the <strong>DEC_SPD_By_Location_FC</strong> test or follow the <strong>DEC_SPD_Unsupported_Parameters_FC</strong> test.</p><p>For efficiency, this test may be conducted with  multiple DOCs if supported, and if desired. </p><p>If multiple patients are supported by the DOF, this  test will be conducted with two patients. If either system does not support  multiple patients, you must make this known to your test partners and the  project manager. </p><p>A DOC requests  data as described below, and the DOF responds with the requested data. Messages  include at least three physiological parameters for each type of system  supported (e.g., if the systems support bedside monitor and ventilator data, at  least three parameters are required for each). Parameters are selected at the  discretion of the participating systems.. Parameters include sampled continuous  parameters (e.g., heart rate) and <em>should  include</em> relatively infrequently measured parameters (e.g., NIBP) if these  are supported. </p><p> The data source will send messages continuously  before and during each step of this test. The steps are:</p><ol>     <li>If  the DOC is receiving messages it first sends a request to terminate all  transmissions.</li>     <li>The  DOC requests only one set of data for each patient identified for the company.  The complete set for each patient may require more than one message in order to  obtain a snapshot of continuous data (e.g., a single value for heart rate) and  the next value for intermittent data (e.g., NIBP). If a second DOC is involved,  it should delay testing until the first DOC is finished its test.</li>     <li>DOC(s)  terminate the subscriptions.</li> </ol><table border="1" cellspacing="0" cellpadding="0">     <tr>      <td width="120" valign="top"> <p align="center"> <strong>System</strong> </p> </td>     <td width="90" valign="top"> <p align="center"> <strong>MRN</strong> </p> </td>     <td width="60" valign="top"> <p align="center"> <strong>First Name</strong> </p> </td>     <td width="66" valign="top"> <p align="center"> <strong>Last Name</strong> </p> </td>     <td width="142" valign="top"> <p align="center"> <strong>Location</strong> </p> </td>     <td width="79" valign="top"> <p align="center"> <strong>Mother&rsquo;s Maiden Name</strong> </p> </td>     <td width="73" valign="top"> <p align="center"> <strong>DOB</strong> </p> </td>     </tr>     <tr>      <td width="120" valign="top"> <p>&nbsp; B Braun (DOF)</p> </td>     <td width="90" valign="top"> <p>BB2008001</p> </td>     <td width="60" valign="top"> <p>Albert</p> </td>     <td width="66" valign="top"> <p>Brooks</p> </td>     <td width="142" valign="top"> <p>3 West ICU - 31-1</p> </td>     <td width="79" valign="top"> <p>Coburn</p> </td>     <td width="73" valign="top"> <p>1/1/61</p> </td>     </tr>     <tr>      <td width="120" valign="top"> <p>&nbsp; B Braun (DOF)</p> </td>     <td width="90" valign="top"> <p>BB2008007</p> </td>     <td width="60" valign="top"> <p>Irma </p> </td>     <td width="66" valign="top"> <p>Kwan</p> </td>     <td width="142" valign="top"> <p>Emergency - ED4-2</p> </td>     <td width="79" valign="top"> <p>Lee</p> </td>     <td width="73" valign="top"> <p>1/1/62</p> </td>     </tr>     <tr>      <td width="120" valign="top"> <p>&nbsp; Philips (DOF)</p> </td>     <td width="90" valign="top"> <p>PH2008001</p> </td>     <td width="60" valign="top"> <p>Adrian</p> </td>     <td width="66" valign="top"> <p>Brooks</p> </td>     <td width="142" valign="top"> <p>3 West ICU &ndash; 34-1</p> </td>     <td width="79" valign="top"> <p>Coburn</p> </td>     <td width="73" valign="top"> <p>1/1/64</p> </td>     </tr>     <tr>      <td width="120" valign="top"> <p>&nbsp; Philips (DOF)</p> </td>     <td width="90" valign="top"> <p>BB2008007</p> </td>     <td width="60" valign="top"> <p>Ira</p> </td>     <td width="66" valign="top"> <p>Kwan</p> </td>     <td width="142" valign="top"> <p>Emergency - ED4-2</p> </td>     <td width="79" valign="top"> <p>Lee</p> </td>     <td width="73" valign="top"> <p>1/1/65</p> </td>     </tr>     <tr>      <td width="120" valign="top"> <p>&nbsp; Spacelabs (DOF)</p> </td>     <td width="90" valign="top"> <p>SP2008001</p> </td>     <td width="60" valign="top"> <p>Arthur</p> </td>     <td width="66" valign="top"> <p>Brooks</p> </td>     <td width="142" valign="top"> <p>3 West ICU &ndash; 35-1</p> </td>     <td width="79" valign="top"> <p>Coburn</p> </td>     <td width="73" valign="top"> <p>1/1/65</p> </td>     </tr>     <tr>      <td width="120" valign="top"> <p>&nbsp; Spacelabs (DOF)</p> </td>     <td width="90" valign="top"> <p>BB2008007</p> </td>     <td width="60" valign="top"> <p>Ima</p> </td>     <td width="66" valign="top"> <p>Kwan</p> </td>     <td width="142" valign="top"> <p>Emergency - ED4-2</p> </td>     <td width="79" valign="top"> <p>Lee</p> </td>     <td width="73" valign="top"> <p>1/1/66</p> </td>     </tr> </table><H3>Evaluation</H3><p>Project manager or monitor will observe the displays provided by the DOF and DOC. The physiologic parameters and their measurement values, including the time of parameter measurement will be compared. The type of display/documentation by the DOF and DOC is not constrained, it need only be clinically useful.</p><p>Success: If the patient, the minimum number of parameters, measurements and time of measurement sent by the DOF matches those received by the DOC in each of the three steps above, that will indicate the test has been successful.</p> 