<H3>Special Instructions</H3><p> <p>When entering results for this test in Kudu, the Image Display should initiate an instance of this test <strong>when the results are complete</strong>.  This allows you to also give credit to the Acquisition Modality or Evidence Creator which supplied the waveforms/images/reports without locking that system out of other testing while the Image Display is working on the rendering.</p> <p>The Image Display may use the Connectathon CENTRAL_ARCHIVE as a source of different vendors' waveforms, images and reports</p> <p>Image Display please use the <em><a href=system_con_object.php>Objects Information</a></em> page to find out what your peers have uploaded on the central archive </p> </p>  <H3>Description</H3> <p>The test verifies that the Image Display can retrieve and display a variety of waveforms, image sets, DICOM SRs and Procedure Logs from various vendors.</p>  <H3>Evaluation</H3> <p>On the<strong> Image Display </strong> </p> <ul> <li>verify that waveforms and images from each Modality vendor can be accurately stored and displayed.  The Connectathon Monitor may ask a representative of the originating Modality to verify the accuracy of the rendering of its study on the Image Display.</li> <li>For waveform objects, verify: <ul> <li>Each waveform is labeled with the lead name.</li> <li>The following values must be displayed: <ul> <li>Voltage scale</li> <li>Time scale</li> <li>Frequency content</li> <li>Acquisition time</li> <li>Originality  eg.  Original (rhythm) or derived (representative beat)</li> <li>Protocol name</li> <li>Protocol stage</li> <li>Patient state</li> </ul> </li> </ul> </li> <li>Verify the ability to display DICOM Encapsulated PDF</li> <li>The Connectathon Monitor should observe its ability to interpret/render the objects (derived images, SRs, procedure logs, or measurements, ie Evidence Documents) from the Evidence Creator actors. The Connectathon Monitor may ask a representative of the originating Evidence Creator to verify the accuracy of the rendering of its data, or observe the originals at the Evidence Creator and spot-check that they are displayed correctly and completely. </li> <li>Verify the ability to display the value of <b>Series Description</b> for each series displayed (RAD-TF:4.16.4.2.3)</li> <li>Accuracy of the NM display is not verified here. It is verified within the scope of the Nuclear Medicine Image Profile Connectathon tests.</li> </ul> 