<h3>Special Instructions</h3> <p>DEC_Infusion_Pump is defined in the context of DOR/DOC pairs where the DOR is an Infusion Pump. If your DOR is not an Infusion Pump, do not perform this test.</p>  <h3>Description</h3> <p>DORs will send messages with reasonable physiologic values for supported parameters. Each message shall contain the required number of parameters that the receiving DOC can process and display or document, and may send any number of additional parameters. DORs are responsible for coordinating with their DOC partners in advance of the Connectathon to assure that the DOCs are able to receive at least the minimum three parameters required for a given test. Parameter values should be realistic for a patient in a cardiac ICU bed or on an OR table.</p>  <p>At least four parameters will be be sent for each patient; at least three of which can be processed by the receiving DOC. Minimum parameters are selected from this list:</p>  <table>  <tr><th>Parameter</th><th>Observation Sub-ID</th><th>Label</th><th>Units (UCUM)</th></tr>  <tr><td>MDC_ATTR_PT_WEIGHT</td><td>w.0.0.2</td><td>Patient Weight</td><td>kg</td></tr>  <tr><td>MDC_PUMP_MODE</td><td>w.1.1.101</td><td>Pump Mode</td><td>n/a</td></tr>  <tr><td>MDC_PUMP_STAT</td><td>w.1.1.102</td><td>Pump Status</td><td>n/a</td></tr>  <tr><td>MDC_FLOW_FLUID_PUMP</td><td>w.1.1.103</td><td>Total Rate</td><td>mL/h</td></tr>  <tr><td>MDC_FLOW_FLUID_PUMP</td><td>w.1.2.201</td><td>Primary Channel Rate</td><td>mL/h</td></tr>  <tr><td>MDC_FLOW_FLUID_PUMP</td><td>w.1.3.301</td><td>Secondary Channel Rate</td><td>mL/h</td></tr>  <tr><td>MDC_VOL_FLUID_TBI_REMAIN</td><td>w.1.2.202</td><td>Primary Channel VTBI</td><td>mL</td></tr>  <tr><td>MDC_VOL_FLUID_TBI_REMAIN</td><td>w.1.3.302</td><td>Secondary Channel VTBI</td><td>mL</td></tr>  <tr><td>MDC_TIME_PD_REMAIN</td><td>w.1.2.203</td><td>Primary Channel Time Remaining</td><td>min</td></tr>  <tr><td>MDC_TIME_PD_REMAIN</td><td>w.1.3.303</td><td>Secondary Channel Time Remaining</td><td>min</td></tr>  <tr><td>MDC_DRUG_NAME_TYPE</td><td>w.1.2.204</td><td>Primary Channel Drug Name</td><td>n/a</td></tr>  <tr><td>MDC_DRUG_NAME_TYPE</td><td>w.1.3.304</td><td>Secondary Channel Drug Name</td><td>n/a</td></tr>  <tr><td>MDC_CONC_DRUG</td><td>w.1.2.205</td><td>Primary Channel Concentration</td><td>mg/mL</td></tr>  <tr><td>MDC_RATE_DOSE</td><td>w.1.2.206</td><td>Primary Channel Dosage</td><td>ug/kg/min</td></tr> </table>  <h3>Evaluation</h3> <p>Project manager or monitor will observe the displays provided by the DOR and partner DOC. The physiologic parameters and their measurement values, including the time of parameter measurement will be compared. The type of display/documentation by the DOC is not constrained, it need only be clinically useful.</p>  <p>If the minimum number of parameters, measurements and time of measurement sent by the DOR matches those received by the DOC that will indicate the step in the test has been successful. NOTE: At least one message sent by the DOR for both patients will not have been received by the DOC (Communication Link Interrupted And Resumed) and the time recorded by the DOR for the interruption will be compared to the missing data point(s). Note also that the data must match the patient.</p> 