<H3>Special Instructions</H3> <p>For this test, Clinician uses a BCMA that supports multiple pumps and multiple patients to initiate PCD-03 transactions consistent with the steps described below. Vendors will document limitations if their BCMA does not support multiple pumps and multiple patients. </p>  <p>If capable of sending data over associated DOR to DOC transactions, these systems also test with multiple DOCs ' a DOC associated with the IOP if one is present, and a DOC that is not associated with the BCMA's IOP. </p>  <p>Clinician uses BCMA Programmer to enter patient and pump identification and to identify the drug and the delivery parameters. Patient demographics are provided at the Kudu test page PCD_Patient_Demographics. Specific drug and parameters are given in the Kudu test page PIV_Parameter_Tables.  </p>  <H3>Description</H3> <p>Developer prepares the BCMA Programmer to provide patient and pump identification and to identify the drug and the delivery parameters. Patient demographics are provided at the Kudu test page PCD_Patient_Demographics. Specific drug and parameters are given in the Kudu test page PIV_Parameter_Tables, http://ihe-kudu.wustl.edu/na2009/connectathon_test.php.  </p>  <p>The pumps should run long enough to make it obvious that some fluid was delivered. If the IOP and IOC implement the DEC PCD-01 profile, that data may be used to help document what happened. </p>  <p>The IOP requests delivery of drugs for two patients on each of two or more IOCs. Only one pump is used for each patient. I.e., each IOC supports two or more patients, each with one pump. The DORs associated with the IOCs send messages to the DOC associated with the IOP and to at least one other DOC. </p>  <p>Parameters should be sent from actual devices (preferred), simulators or, if necessary from parameter tables provided by the Project Manager (least desirable). </p>  <H3>Evaluation</H3> <p>Request a monitor validate performance.  Be prepared to demonstrate that the record of performance matches the record of the performance requested: the IOC received and, with support of the clinician, programmed the right pump to deliver the right drug, right amount, at the right flow rate via the right site to the right patient. The DOCs' pump data should support the performance record. </p>  <p>Project manager or monitor will observe the displays provided by the sending and receiving system(s) and the pumps. The patients' names and/or medical record numbers (MRNs), physiologic and other parameters, their measurement values, and the time of parameter measurement/acquisition and receipt will be compared. </p>  <p>The type of display/documentation by the receiving system is not constrained, it need only be clinically useful. </p> 