<H3>Special Instructions</H3> <b><i>This test should be run on the 1st or 2nd day of the Connectathon to enable Image Displays to do a follow-up test to Q/R these images and evidence objects.</b></i> <H3>Description</H3> The purpose of this test is to verify that an Image Manager can accept and store real world vendor images and evidence objects which are available.  Q/R is tested in another test <H3>Evaluation</H3> Verify that all images and evidence objects from several Acquisition Modality and Evidence Creator actors in the SWF profile are received at IM under test. Verify the total number of SOP instances stored and total number of studies stored. Verify that the objects are stored as full-fidelity objects (ie., no DICOM data elements are deleted). <p> Image Managers must support storage of multiple SOP classes (RAD TF-2:4.8.4.1.3.1):  </p> <pre> SOP Class UID 	SOP Class Name  1.2.840.10008.5.1.4.1.1.1  Computed Radiography Image Storage  1.2.840.10008.5.1.4.1.1.2  CT Image Storage  1.2.840.10008.5.1.4.1.1.4 MR Image Storage  1.2.840.10008.5.1.4.1.1.20 Nuclear Medicine Image Storage  1.2.840.10008.5.1.4.1.1.128 Positron Emission Tomography Image Storage  1.2.840.10008.5.1.4.1.1.481.1 RT Image Storage  1.2.840.10008.5.1.4.1.1.7 Secondary Capture Image Storage  1.2.840.10008.5.1.4.1.1.6.1 Ultrasound Image Storage  1.2.840.10008.5.1.4.1.1.3.1 Ultrasound Multi-frame Image Storage  1.2.840.10008.5.1.4.1.1.12.1 X-Ray Angiographic Image Storage  1.2.840.10008.5.1.4.1.1.12.2 X-Ray Radiofluoroscopic Image Storage  1.2.840.10008.5.1.4.1.1.1.1 Digital X-Ray Image Storage ' For Presentation  1.2.840.10008.5.1.4.1.1.1.1.1 Digital X-Ray Image Storage ' For Processing  1.2.840.10008.5.1.4.1.1.1.2 Digital Mammography Image Storage ' For Presentation  1.2.840.10008.5.1.4.1.1.1.2.1 Digital Mammography Image Storage ' For Processing  1.2.840.10008.5.1.4.1.1.1.3 Digital Intra-oral X-Ray Image Storage ' For Presentation  1.2.840.10008.5.1.4.1.1.1.3.1 Digital Intra-oral X-Ray Image Storage ' For Processing  1.2.840.10008.5.1.4.1.1.77.1.1 VL Endoscopic Image Storage  1.2.840.10008.5.1.4.1.1.77.1.2 VL Microscopic Image Storage  1.2.840.10008.5.1.4.1.1.77.1.3 	VL Slide-Coordinates Microscopic Image Storage  1.2.840.10008.5.1.4.1.1.77.1.4 	VL Photographic Image Storage  </pre>