<H3>Special Instructions</H3><p> <p>This test is run by Image Display systems which support  any of the cardiology options in the Evidence Documents profile:</p> <ul>     <ul>      <li>Cath Evidence option</li>      <li>Echo Evidence option</li>      <li>CTA/MRA Evidence option</li>      <li>Stress Test Evidence option </li>     </ul> </ul> <p>In order to run this test, Evidence Creator actors which  support these options must complete the tests in which they store their evidence  documents (SRs) to Image Manager systems.&nbsp;  These tests are named ED_Creator_PACS_*_Option.</p> </p><H3>Description</H3><p>In this test, the Image Display demonstrates its ability  to cardiology evidence documents (SRs) created by various Evidence Creator  systems.</p><p>    The Image Display must be able to render and display all  Templates for the SOP Classes which it accepts.</p><p>    The Image Display system should drive this test.&nbsp; The representative of the Image&nbsp; Display system should identify an Evidence  Creator system which supports the same cardiology option(s) as its system. Ask  the Evidence Creator vendor representative which Image Manager/Image Archive it  has stored its evidence documents (SRs) to.</p><p>    Note 1: The Evidence Creator is a required actor in this  test.&nbsp; It has&nbsp; no transactions to perform, but will be  identified so that it can be given credit for interoperability with the Image  Display who renders its SRs.</p><p>    Note 2:&nbsp; The Image  Manager system is identified as &lsquo;optional&rsquo; for this test because it is needed  to help out with Image Display system, allowing it to Query/Retrieve SRs from  its system.&nbsp; Image Managers are not  required to complete&nbsp; 3 instances of this  test.</p><H3>Evaluation</H3><p>FOR ALL CARDIOLOGY OPTIONS (Cath Evidence, Echo Evidence,  CTA/MRA Evidence, Stress Test Evidence:</p><ul>     <li>The Connectathon Monitor should observe &lsquo;live&rsquo;  the ability of the Image Display to query and retrieve SRs from an Image  Manager system (not the Central Archive).</li>     <li>The Connectathon Monitor should observe the  Image Display and its ability to interpret/render the Evidence Documents from  the Evidence Creator. All SOP Classes and Templates which the Evidence Creator  supports should be displayed.</li>     <li>The Connectathon Monitor should observe the original  report at the Evidence Creator (the creator of the SR), the Image Manager, or  another Image Display and spot check the Image Display/Report creator to verify  the contents of the Structured Report are displayed <u>correctly</u> and <u>completely</u> </li>     <li>The Image Display must demonstrate its ability  to show abnormal measurements with &ldquo;readily visible emphasis&rdquo;, (e.g., font,  bold, text or background color, specialized window area, etc.).</li> </ul><p>For the <strong>CTA/MRA Evidence </strong>option:</p><ul>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object for a content item with value type  of IMAGE.</li>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object with rendering of spatial  coordinates for a content item with value type of SCOORD.</li> </ul><p>For the <strong>Stress Test Evidence </strong>option:</p><ul>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object for a content item with value type  of IMAGE.</li>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object with rendering of temporal or  spatial coordinates for a content item with value type of TCOORD or SCOORD.</li>     <li>The Image Display actor must  demonstrates user control for rendering of ECG ST elevation or depression  measurements either in units of mV or in units of mm (based on conventional  scaling of 0.1 mV/mm). </li>     <li>For SR documents  encoded using Template x3300, the Image Display must render the contents of the  Template x3304 Stress Test Measurement Groups (subsidiary to a Template x3303  Stress Test Phase Data CONTAINER) <strong><u>in  a tabular format</u></strong>. </li> </ul> 