<H3>Special Instructions</H3><p> <p>This test is run by Image Display systems which support  any of the cardiology options in the Evidence Documents profile:</p> <ul>     <ul>      <li>Cath Evidence option</li>      <li>Echo Evidence option</li>      <li>CTA/MRA Evidence option</li>      <li>Stress Test Evidence option </li>     </ul> </ul> <p>In order to run this test, Acquisition Modality actors  which support these options must complete the tests in which they store their  evidence documents (SRs) to Image Manager systems.  These tests are named  ED_Modality_PACS_*_Option.</p> <p>Image Display please use the <em><a href=system_con_object.php>Objects Information</a></em> page to find out what your peers have uploaded on the central archive </p>  </p><H3>Description</H3><p>In this test, the Image Display demonstrates its ability  to cardiology evidence documents (SRs) created by various Acquisition Modality  systems.</p><p>    The Image Display must be able to render and display all  Templates for the SOP Classes which it accepts.</p><p>    The Image Display system should drive this test.  The representative of the Image  Display system should identify an Acquisition  Modality System which supports the same cardiology option(s) as its system. Ask  the Modality vendor representative which Image Manager/Image Archive it has  stored its evidence documents (SRs) to.</p><p>    Note 1: The Acquisition Modality is a required actor in  this test.  It has  no transactions to perform, but will be  identified so that it can be given credit for interoperability with the Image  Display who renders its SRs.</p><p>    Note 2:  The Image  Manager system is identified as 'optional' for this test because it is needed  to help out with Image Display system, allowing it to Query/Retrieve SRs from  its system.  Image Managers are not  required to complete  3 instances of this  test.</p><H3>Evaluation</H3><p>FOR ALL CARDIOLOGY OPTIONS (Cath Evidence, Echo Evidence,  CTA/MRA Evidence, Stress Test Evidence:</p><ul>     <li>The Connectathon Monitor should observe 'live'  the ability of the Image Display to query and retrieve SRs from an Image Manager  system (not the Central Archive).</li>     <li>The Connectathon Monitor should observe the  Image Display and its ability to interpret/render the Evidence Documents from  the Acquisition Modality. All SOP Classes and Templates which the Acquistion  Modality supports should be displayed.</li>     <li>The Connectathon Monitor should observe the  original report at the Acquisition Modality (the creator of the SR), the Image  Manager, or another Image Display and spot check the Image Display/Report  creator to verify the contents of the Structured Report are displayed <u>correctly</u> and <u>completely</u> </li>     <li>The Image Display must demonstrate its ability  to show abnormal measurements with 'readily visible emphasis', (e.g., font,  bold, text or background color, specialized window area, etc.).</li> </ul><p>For the <strong>CTA/MRA Evidence </strong>option:</p><ul>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object for a content item with value type  of IMAGE.</li>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object with rendering of spatial  coordinates for a content item with value type of SCOORD.</li> </ul><p>For the <strong>Stress Test Evidence </strong>option:</p><ul>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object for a content item with value type  of IMAGE.</li>     <li>The Image Display must demonstrate that it can  display a hyperlink to a referenced object with rendering of temporal or spatial  coordinates for a content item with value type of TCOORD or SCOORD.</li>     <li>The Image Display actor must  demonstratea user control for rendering of ECG ST elevation or depression  measurements either in units of mV or in units of mm (based on conventional  scaling of 0.1 mV/mm). </li>     <li>For SR documents  encoded using Template x3300, the Image Display must render the contents of the  Template x3304 Stress Test Measurement Groups (subsidiary to a Template x3303  Stress Test Phase Data CONTAINER) <strong><u>in  a tabular format</u></strong>. </li> </ul> 