<H3>Special Instructions</H3> <p>Test requires that the IOC be turned off or power removed before beginning test sequence. The IOP and pump are available.  </p>   <H3>Description</H3> <p>For this test, Clinician uses a BCMA to initiate PCD-03 transactions consistent with the following steps. Vendors will document the pump is on, the IOP is available, and the IOC is not. </p>  <p>Clinician uses BCMA Programmer to enter patient and pump identification and to identify the drug, its concentration, and volume to be delivered. Pump is to deliver the same drug as in ____ test at half the flow rate; the concentration remains the same. Pump remains on its assigned patient, starting when user verifies command. </p>  <ol>  <li>Clinician verifies that IOP and pump were available.</li>  <li>IOP times out a t network level. Clinician does not receive confirmation message. If a message is provided to the clinician by the IOP this will be shown to the monitor. </li> <li>Clinician restores power to the IOC. IOP either retransmits exiting request(s) or clinician must initiate new request. </li>  </ol>  <H3>Evaluation</H3> <p>Monitor will observe that the network timed out, and an error message if any is displayed by the programmer. Monitor will observe sequence of events following restoration of power to the IOC. This includes transmission of the request to IOC and confirmation from the IOC received by the IOP. </p>